In-use effect of electrolysed water on transcutaneous oxygen sensorsStephanie J. Dancer A D , Julie Mallon A , Rebecca Murphy B and Cliff Murch C
A Department of Microbiology, Hairmyres Hospital, NHS Lanarkshire, East Kilbride, Lanarkshire G75 8RG, Scotland.
B Department of Podiatry, Hairmyres Hospital, NHS Lanarkshire, East Kilbride, Lanarkshire G75 8RG, Scotland.
C Department of Radiology, Hairmyres Hospital, NHS Lanarkshire, East Kilbride, Lanarkshire G75 8RG, Scotland.
D Corresponding author. Email: firstname.lastname@example.org
Healthcare Infection 20(4) 141-144 https://doi.org/10.1071/HI15017
Submitted: 8 August 2015 Accepted: 7 September 2015 Published: 1 October 2015
Reusable clinical equipment should be decontaminated between patients in order to reduce risk of pathogen transmission. Manufacturers are obliged to offer advice regarding decontamination but occasionally insufficient guidance is provided. Tissue oxygen sensors are reusable, costly and vulnerable to disinfectants. This pilot study describes an in-use protocol using neutral electrolysed water aimed at eliminating infection risk during transcutaneous oxygen monitoring of diabetic feet. Sensor components were screened for microbial contamination before, during, and after monitoring healthy and diabetic volunteers. Exposure to electrolysed water removed original skin commensals and alleviated the risk of transmitting microbial flora without affecting test results. The product is non-toxic, inexpensive and may be useful for decontaminating a wide range of sensitive clinical equipment.
References Chao CY, Cheing GL. Microvascular dysfunction in diabetic foot disease and ulceration. Diabetes Metab Res Rev 2009; 25 604–14.
| Microvascular dysfunction in diabetic foot disease and ulceration.CrossRef | 1:CAS:528:DC%2BD1MXhtlyqsbvM&md5=0b3e5ef919e9577c204c0724ad9ce5adCAS | 19681035PubMed |
 Mathieu D, Mani R. A review of the clinical significance of tissue hypoxia measurements in lower extremity wound management. Int J Low Extrem Wounds 2007; 6 273–83.
| A review of the clinical significance of tissue hypoxia measurements in lower extremity wound management.CrossRef | 18048873PubMed |
 Franklin ML. Transcutaneous measurement of partial pressure of oxygen and carbon dioxide. Respir Care Clin N Am 1995; 1 119–31.
| 1:STN:280:DyaK1c%2Fks1CnsA%3D%3D&md5=779845642aac857438c4920fad9b8e8fCAS | 9390854PubMed |
 Rutala WA, Weber DJ. Healthcare Infection Control Practices Advisory Committee. Guidelines for disinfection and sterilization in healthcare facilities, 2008. Centers for Disease Control and Prevention; 2008. Available at: http://www.cdc.gov/hicpac/Disinfection_Sterilization/acknowledg.html [verified September 2015]
 Thorn RMS, Lee SWH, Robinson GM, Greenman J, Reynolds DM. Electrochemically activated solutions: evidence for antimicrobial efficacy and applications in healthcare environments. Eur J Clin Microbiol Infect Dis 2012; 31 641–53.
| 1:CAS:528:DC%2BC38Xlt1ejtbY%3D&md5=16e868d10e74bf1b03ad2e92e49853bdCAS |
 Pintaric R, Matela J, Pintaric S. Suitability of electrolyzed oxidising water for the disinfection of hard surfaces and equipment in radiology. J Environ Health Sci Eng 2015; 13 6
| Suitability of electrolyzed oxidising water for the disinfection of hard surfaces and equipment in radiology.CrossRef | 25642329PubMed |
 Meakin NS, Bowman C, Lewis M, Dancer SJ. Cleaning efficacy between in-use disinfectant and electrolysed water in an English residential care home. J Hosp Infect 2012; 80 122–7.
| Cleaning efficacy between in-use disinfectant and electrolysed water in an English residential care home.CrossRef | 1:STN:280:DC%2BC387mtFKgsg%3D%3D&md5=73062c40d1eeb6505574d781e9f222d4CAS | 22196853PubMed |
 Stewart M, Bogusz A, Hunter J, Devanny I, Yip B, Reid D, Robertson C, Dancer SJ. Evaluating use of neutral electrolyzed water for cleaning near-patient surfaces. Infect Control Hosp Epidemiol 2014; 35 1505–10.
| Evaluating use of neutral electrolyzed water for cleaning near-patient surfaces.CrossRef | 1:STN:280:DC%2BC2Mzgs1Wrug%3D%3D&md5=7353b5973734cb74dd6f55357ce6ba34CAS | 25419773PubMed |
 Reichert M, Young J. Sterilization technology for the health care facility. 2nd edn. Jones & Bartlett Learning; 1997.
 Fertelli D, Cadnum JL, Nerandzic MM, Sitzlar B, Kundrapu S, Donskey CJ. Effectiveness of an electrochemically activated saline solution for disinfection of hospital equipment. Infect Control Hosp Epidemiol 2013; 34 543–4.
| Effectiveness of an electrochemically activated saline solution for disinfection of hospital equipment.CrossRef | 23571379PubMed |